Message 1 of The help file presents application programming examples, a cross-reference between instrument commands and driver functions, and detailed documentation of each function, with examples. There are several advantages to using the dll form of the driver. Did you get this working? A small update on the setup definition I use: I would like to know if there is any VI available for 2 terminal devices in this case on wafer diodes so that I can control directly the equipment from LabView and be able to read out the measured data directly from the HP via LabView? Most Active Software Boards:
|Date Added:||26 November 2006|
|File Size:||53.97 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Message 6 of Sorry, I’ve been on vacation.
Solved: LabView VI for HP : diode measurement IV – Discussion Forums – National Instruments
I can now input data from Labview to HP The rest of the channels are not assigned, as now I only try one port measurement, to see if I can take 4516b data from the HP A small update on the setup definition I use: I will at least try that, I tried working with two other drivers as well and somehow sampling is way harder than sweeping. Hello, it’s me again with my very old and troubled semiconductor analyzer. After this I can see the cursor on the bottom of the I V graph on the monitor of the sliding from left to right when settinp a Sweep of Voltage of SMU1 from 0 to 1 V, pts.
Message 4 of 6.
Agilent 4156 Analyzer
I have a very old VI with a very old driver. Two elements of the standard do not apply to the instrument since it is not a VXI instrument. Did you get this working? I will write Keysight, even though this product I believe to be discontinued, maybe they can offer me some support. Labfiew on HP Will keep you updated if it works.
Auto-suggest helps you quickly narrow down your search results labvies suggesting possible matches as you type. A list of contact information for other countries is available on the Agilent Technologies website at http: Message 2 of 6.
The SCPI commands look similar for both, but there are probably subtle differences that are the result of new features in the Agilent C.
Agilent Technologies / Keysight Technologies 4156B
Second, there is no knowledge base file, which is primarily a description of a VXI board. I am new to the forum and also new to LabView. Do you think somethings wrong with the following?
Message 7 of To load the dll you must include the file hpb. It’s alright, I now have a code that works perfectly! This lavbiew on the B, right? It includes a “Function Panel”.
The VI instrument driver works. I would like to know if there is any VI available for 2 terminal devices in this case on wafer diodes so that I can control directly the equipment from LabView and be able to read out the measured data directly from labgiew HP via LabView?
I will try nevertheless to find a solution.
You must also provide an include path for hpb. I managedto get communication between the equipment HP and LabView. Message 9 of However I get this error and nothing is showing up not on the equipment neither in the VI interface in LabView.
Message 2 of